|Stations:||6 Groups||Load Current:||40A/ 12V|
|Power Up Time:||0 ~ 99 Min Adjustable||Tensile Height:||254 Mm (10 "), 178 Mm (7") 2 Stalls|
UL817 Volume Test Machine,
UL817 Mask Test Machine,
VDE0620 Volume Test Machine
Respiratory Protective Mask Carbon Dioxide (CO2) Volume Test Machine
The mask dead space detection device is used to detect the dead space of the protective mask, that is, the volume fraction of CO2 in the inhaled gas.
Comply with the standards:
The main components of the test device include: test head mold, mechanical artificial lung with solenoid valve, connecting pipe, flowmeter, CO2 gas analyzer and control system.
The test device is driven by the low-speed motor of the detection device through the control system to drive the artificial lung to make reciprocating motion to simulate the actual breathing process of human beings. Pushing is equivalent to inhaling, and returning is equivalent to exhaling. In terms of this process, the upper part of the cupboard is completed. At the same time, a certain amount of high concentration CO2 was injected into the breath, and the whole process of respiration was detected by CO2 gas analyzer.
|Respiration simulator simulation frequency modulation||10-40 times/min|
|Adjusting range of simulated tidal volume||0.5～3.0L/min|
|The volume fraction of carbon dioxide gas source||5.0±0.1%|
|Carbon dioxide flow meter range||not less than 40L/min|
|Carbon dioxide analyzer range||no less than 12%, accuracy no less than 0.1%|
|Ventilated electric fan||wind speed 0.5m/s|
|Respiratory rate||20 times/min|
|Respiratory tidal volume||1.5L|
|Power supply||Power supply|
Real-time automatic calculation and display of the volume fraction of CO2 in the inhaled gas.
Contact Person: Sophia Su